Cyclic Triaxial testing
As my load cell showed an initial value of around 300N, zeroing was done. After zeroing, a negative value (-5 N) is read. Can I proceed with a negative value for further calibration? Or else how to fix this issue? It is given in the manual that we should not proceed with a negative value or tension load.